What kind of ventilators can be used with the Sedaconda ACD?+
Only use CE‑approved ventilators complying with applicable requirements, including standard ISO 80601‑2‑12. Sedaconda ACD can be used with all conventional ventilator modes except oscillator mode for intubated patients. Use ventilator circuits compatible with anaesthetic agents. Use only with ventilators with an accessible exhaust.
How long can the Sedaconda ACD be used for?+
Sedaconda ACD is intended for single use only and needs to be replaced every 24 hours, or earlier if needed. Example, if unexpected events occur such as sudden blockage of the airways due to excessive secretion, etc.
Can N2O be used in the gas mixture with the Sedaconda ACD?+
Sedaconda ACD is not approved for use with N2O.
What type of Syringe pumps should I use?+
The syringe pump should be CE marked and programmable for delivery of pharmaceutical drugs and should have settings for Sherwood Monoject or Becton Dickinson Plastipak 50/60 ml syringes.
Can other syringes be used with the Sedaconda ACD?+
No, the Sedaconda Syringe has been tested and validated for use with Isoflurane or Sevoflurane. The connector on the syringe is unique, and can only be connected to the filling adapters provided by Sedana Medical and the Sedaconda ACD agent line.
Which anaesthetic agents can be used with Sedaconda ACD?+
Either Isoflurane or Sevoflurane can be used.
Can a nebulizer be used with Sedaconda ACD?+
It is possible to use Jet/Ultrasound/Mesh nebulizer with Sedaconda ACD. The nebulizer should be connected between the patient and Sedaconda ACD. Ultrasonic nebulisers are preferable as they do not add extra airflow. If a jet nebuliser is connected it may be necessary to increase the syringe pump rate to compensate for the extra flow from the nebuliser. Extra flow may decrease the gas concentration. Please refer to the relevant manufacture for compatibility with volatile anaesthetics.
Note. Repeated nebulisations may increase the flow resistance of the Sedaconda ACD. Pay attention to signs of occlusions. Always consider the increased dead space when connecting extra items.
Can pressure support ventilation be used with the Sedaconda ACD?+
Yes, but there may be some changes in concentration. Normal fluctuations during pressure support ventilation do not change the gas concentration in any significant way.
Is it possible to temporary stop the treatment?+
Yes, the syringe pump needs to be stopped and the gas sampling line disconnected. First disconnect the Sedaconda ACD from the Y-piece and then from the ET-tube. It is vitally important to disconnect or stop the gas monitor as it draws between 50-200 ml/min of gas. When reconnecting the patient please follow the Instructions for Use.
Can the Sedaconda ACD be used with Active Humidification?+
Yes, but to use active humidification, the Sedaconda ACD needs to be place at the inspiratory limb. See IFU for Ventilator/Alternative placement. Active humidification cannot be used with patient/standard placement.
What is a MAC and Fet% value?+
The minimal alveolar concentration (MAC) is the alveolar concentration of an inhaled anaesthetic at one atmosphere that prevents skeletal muscle movement in response to a noxious stimulus in 50% of patients. MAC represents a relative measure of the level of anaesthesia. It can be used to compare inhaled anaesthetics and is a useful index of anaesthetic potency. MAC also corresponds to Fet% and viceversa.
What type of gas sampling line can be used?+
Always refer to the gas analyser manual regarding which sampling line to use. Connecting the incorrect type or length of line may cause incorrect Fet/MAC readings from the gas analyser.
Why is the Fet value higher than the Fi value?+
The highest gas concentration occurs at the start of the inspiration cycle when the recycled gas is mixed with the fresh gas from the Sedaconda ACD evaporator. However this represents a short peak and the concentration decreases rapidly to its lowest point at the end of inspiration. Most monitors show a Fi value which is measured at the end of inspiration thus showing a low value when using the Sedaconda ACD. The expiratory values are measured at the end of expiration which gives the value of alveolar concentration.